I put the thermocouple into operation.
The Basic Configuration with Thermocouple Configuration has been made. Only the Can transfer rate was changed to 250kbit / s.
The values are displayed using the canalyzer.
The values are too high. Type K sensor is used. Where can I set the conversion or units?
The MU-TC1 Thermocouple sends data in 1/16 degrees (see manual p.26).
So you may either set a scale of 16 in the device so it sends full degree.
Or you may apply a factor 16 in your visualization system.
A tutorial (in german) is found on the product DVD:
\Tools\Thermocouple Configuration\shipping config\Konfigurieren des MU-TC1.pdf
See exercises 1.x (starting p.8 onward).